Overvolted multimeter? Answered
Simple question. (or maybe not)
I have a digital multimeter I bought a while ago that I think I accidentally put too much voltage through. The voltmeter function on it is now broken. Whenever it's turned on, it reads "156 VDC" even though no voltage is being applied. (however, when I do apply voltage, the meter does detect it.) All other functions seem to be working correctly, but I'm bummed because the voltmeter is what I use the most.
Is there any way to "reset" a multimeter? if not, is there anything I could try to fix it?
Thanks in advance.