Intro: Awesome Hack for Powered Wheelchairs - Automatic Controller Mount
This was a project for Principles and Practices of Assistive Technology (PPAT) in Fall 2016 at MIT. We made a motorized wheelchair controller mount for our client, Lisa. She lives at the Boston Home, a center for people with progressive neurological diseases, and has Multiple Sclerosis.
With her old controller mount, she would often bump into sinks and tables because the controller was difficult to push out of the way and jutted out, preventing her from getting as close as she wanted. She wanted a method to move the mount out of the way easily so that she could access sinks and tables. Our solution was to create a motorized controller mount, which uses a DC Gear Motor and a microcontroller (Arduino UNO) so that Lisa can move the mount with the push of a button. One button press moves the mount away from her, and another press moves it back in front of her.
We understand that this is not just a problem face by Lisa alone. A large number of powered wheelchair users face this and similar problems all over the world. Hence, we created this instructable with open sourced designs for anyone who wants to replicate the project. You can find all the relevant documentation for this project on the Wiki page, including code and design files. In this instructable we will try to show you the steps for creating the same.
According to Lisa, "it works just like I'd imagined."
You can watch a video of our device in action here
Authors: Suma Anand, Abhinav Gandhi, Nora Kelsall, Mark Vrablic
Step 1: Parts Required
We ensured that the parts used for our prototype were common and easily available. Here is a comprehensive list of all the parts with their specifications. We have also added online links wherever possible.
1” Aluminum 6061 Extruded Bar stock
Scrap piece of steel sheet metal 1.1 - 1.2mm thick
Cytron 12V 17RPM 194.4oz-in Spur Gearmotor
Motor mount bracket
Screws and nuts - multiple sizes
set screw: 8-32 size and thread
Heavy duty limit switch
Arduino UNO(we used this as we had one lying around. You can use a smaller board such as arduino nano if you like)
L293D motor driver IC
LM2596 buck converter IC
LM7805 voltage regulator IC
Step 2: Manufacturing the Mechanical Parts
There are two mechanical parts that need to be manufactured. One out of steel sheet metal and another from the extruded aluminium bar.
The CAD files for the two parts are available for download.
Process for machining the aluminium part -
The aluminum part was machined from 1” Aluminum 6061 Extruded Bar stock. First, the stock was milled to create a 1”x1”x1” cube. A ½” end mill was used to cut a centered .52” wide groove, .375” deep to fit around the bar of the joystick mount. A .325” deep, hole ½” in diameter was then drilled through the center of the groove to fit around the nut that holds the bar in place. Subsequently, a hole was drilled to fit around the motor shaft, which is 6mm in diameter. To connect this part to the motor, a set screw hole was drilled and tapped to accommodate a set screw of size and thread 8-32.
Step 3: Fabricating the Electronics
For our final prototype, we used Arduino UNO as the controller along with an L293D motor driver IC, LM2596 DC-DC buck converter module, 7805 voltage regulator IC, and a limit switch. A full schematic of the electronics is available in the images.
The device is powered by the wheelchair battery, which is a 24V, 36Ah lead acid battery. The Arduino controls the motor based on inputs from a button, a limit switch, and inbuilt timing constraints. The LM2596 buck converter IC steps down the 24V supply from the wheelchair battery to 10.5V, which is used to power the Motor and Arduino. The LM7805 voltage regulator IC steps down the 12V supply from the Vin pin of the Arduino to 5V, which is then used to bias the L293D motor driver IC. Pins 12 and 13 of the motor driver are connected to a common ground with the Arduino to ensure that our system has a stable reference voltage level. Pins 11 and 14 provide the output signal to the DC motor, whereas the pins 10 and 15 provide the motor driver with the data signals from the Arduino, which are used to select whether the motor is rotated clockwise or anti-clockwise.
The limit switch sends a signal to the Arduino when pressed against the body of the device. This signifies the swung-out position of the wheelchair arm. To bring the wheelchair arm back to its initial position, we use a timed approach. The motor is turned on for a set amount of time, which was measured, to move the arm forward. You can find a more detailed description of how the Arduino controls the motor on the Code page.
Step 4: Uploading the Code
The code initializes the mount to the forward position based on the limit switch attached to the mount. When the button is first pressed, the motor rotates clockwise and the mount is pushed out of the way until the limit switch fixed under the mount is closed. When the button is pressed again, the motor rotates in the counter-clockwise direction for 2.15 seconds, which brings the joystick mount back to the original position. The code can be downloaded as a .ino file.
Save the file and open it using the Arduino IDE which can be downloaded here.. Connect your Arduino board to the computer via the USB provided. Ensure that the correct board is selected on the 'boards' menu under the 'tools' dropdown. A more clear description on how to upload code to arduino is available here. if you're new to this.
Step 5: Housing and Mounts
Electronic components were housed in a box laser cut from 3.4mm thick black acrylic and held together with epoxy. The plans for the box are shown below, and you can find them here. The inner dimensions are 4 x 7 x 5 (in cm). Alternatively, you can custom design one for laser-cutting here.
The electronics box was taped to the back of Lisa's wheelchair (see images). If you can find a better spot for mounting this please let us know in the comments! Three pairs of wire were fed through the holes in the box - one for the 24V wheelchair battery, and one to the motor, and one to the limit switch. The wires to the motor and limit switch were tied together with zip ties and routed to the front of Lisa's wheelchair, where they were soldered onto the motor and onto the limit switch. Moldable putty and heat shrink tubing was used to protect the solder connections.
Check images for all the details.
Step 6: All Set!
You're all set and ready to go, power it up and test out the shiny new mechanism that you've just put together! Like and share this instructable if you think its helpful. Hope this reaches the niche community of power wheelchair users.
Make sure to check out more resources for people with different abilities -