Introduction: How to Measure the Internal Resistance of a Battery?
* The schematic is taken from other source.
Wondering how well your battery can perform? Don't know how much current you can pull from your battery? Well you can get know to your battery more with a simple experiment. You only need few components to do it.
- your battery(Lipo, lead acid, NiCd, any kind of battery is ok)
- High power resistor (I used 4.7ohm 5W, resistance around 4~10ohm should be fine, but it depends on the battery you are using)
Yep, that's all you need. Since I'm making the post, I need to take the photos and handle the battery as well, so I used battery holder to help me out. However, the results may not be accurate since extra resistance is incurred in the wire itself. And I used a bit longer time to carry out the test, the resistor gets warm and affect the voltage reading.
Lower load resistance can let you get a more accurate results, but by doing so may require less connection period because it drains more current from your battery. Hence your battery voltage will drop more quickly. The load and the battery may get warm (even get hot) and increase their own resistance due to higher temperature, causing more error in the reading.
So make sure you:
- complete the test as quick as possible
- make the period of connection of battery and the resistor as short as possible
- use a reasonable value of resistance, 10ohm should be sufficient for all kind of test.
Step 1: Double Confirm the Resistance of the Resistor
Even thought the value of the resistance is printed, you still need to confirm whether the actual resistance is the same as shown. Most resistor has 10% tolerance in their actual value, but it still depends on the type of resistor you used.
This step is only the easy resistance measurement, so I guess you can do it.
Step 2: Measure the No-load Voltage of the Battery
After that, you can measure the no load voltage of the battery by connecting the test lead to the battery terminals directly. It says "no-load" because the input resistance of the multimeter during the voltage measurement is very high, typically >1Mohm so the effect of current draw can be simply ignored.
Once again, I guess you know how to do it since it is a simple test. So let's go to next step.
Step 3: Measuring the Voltage Across the Load Resistor
Here is the tricky part, you have to complete this test as fast as possible, or else you might not get an accurate reading.
I suggest to connect the resistance across the test lead of the multimeter, then make contact to the battery terminal with the test lead. Make sure your measuring mod is in voltage mode!
As soon as the voltage reading settle down (stop changing very much), record the value and disconnect the battery immediately. The contact should be made no longer than few seconds to prevent the load resistance to draw too much current from the battery and affect the overall reading.
I also found out that the load resistance is getting warm as soon as I disconnect the battery, so I'm expected to get a bad reading.
Once you finished this step, you can proceed to the last step.
Step 4: Finally, Calculation...
Here is the value I get:
- voltage reading across the battery terminal, Va=3.99V
- voltage reading across the load resistor, Vb=3.796V
- load resistance, R=4.7ohm
Since Vb = Va * [ R / (R+r) ], r = internal resistance
Then, internal resistance of the battery can be determined to be around 240mOhm.
Hence, my battery can theoretically discharge the current as high as 4.2V/0.24=17.5A (short circuit maximum current at fully charged state) but it is not good to do so. If higher current output is desired, more batteries should be connected in parallel to reduce the overall internal resistance and hence higher current output capability.
You can repeat the step from measuring the terminal voltage of the battery to get multiple reading, since averaging the reading helps reducing the error. For demonstrative purpose I just get the reading once.
After all, thanks for your ineterest.