How to Detect Concentration of Gas by Using MQ2 Sensor

36,787

19

37

MQ-2 Gas sensor is used to detect the gas level around the area. This sensor able to use as home or factory gas leak monitoring, suitable for gas, butane, propane, methane, alcohol, hydrogen, smoke and other monitoring devices.

Specification:

• Size: 35mm x 22mm x 23mm (length x width x height)
• Main chip: LM393, ZYMQ-2 gas sensors
• Working voltage: DC 5 V
• Characteristics:
1. With a signal output instruction.
2. Dual signal output (analog output, and high/low digital output)
3. 0 ~ 4.2V analog output voltage, the higher the concentration the higher the voltage.
4. Better sensitivity for gas, natural gas, city gas.
5. With a long service life and reliable stability.
6. Rapid response and recovery characteristics.

Teacher Notes

Teachers! Did you use this instructable in your classroom?
Add a Teacher Note to share how you incorporated it into your lesson.

Step 1: Gather Materials

The materials that you need

-LED

Step 3: Set Up the Circuit Above

LCD pins

-Pins D7 , D6 , D5 , D4 connect to arduino digital pins 2,3,4,5 respectively.

-Pins A and K connect to 5V and GROUND respectively

-Pins VSS and VDD connect to Ground and 5V respectively.

LED

-Use as a LED indicator for the calibration process of the MQ2 gas sensor

-Connect to digital pin 13 of Arduino Uno

MQ2 Gas Sensor

-Analog output connect to the A0 of arduino

Potentiometer

-Center pin connect to the V0 of LCD display

-Other two pins just connect to 5V and GROUND respectively

*Note: All of pins above is configured regarding the circuit figure above. You can configure the connection yourself regarding your favourite.

Step 4: Programming the Arduino Uno

As you can see in the graph above , it shows several gases.

The values are plotted on a semilog paper, as you can see the values indicate 100......1000.....10000 ppm.

These values when plotted on a normal graph would represent a curve like an exponential one.

Formula In the Coding

pow(10,( ((log(rs_ro_ratio)-pcurve)/pcurve) + pcurve))

Based on the graph MQ2, the derivation of ppm formula will be:

Y - y1 = m( X - x1)

X= (y-y1)/m + x1

X= (Rs/Ro-y1)/m + x1

log X =(log (Rs/Ro) - y1)/m +x1

X =10^( (log (Rs/Ro) - y1)/m +x1)

*Note: X = ppm on the graph

Y= Rs/Ro

***The derivation formula are same and can be used for other types of MQ gas sensor.

If you have any problem about the coding, you are welcome to ask any question in the comments below or contact us for further details.

1 Person Made This Project!

• 