Introduction: How to Set-Up a Total Station to Conduct Surveying Operations
Every construction project relies on accurate measurements and the Total Station is one of the most important pieces of surveying equipment used in the industry today. From setting building corners to sighting utility lines, this sophisticated instrument provides surveyors with an important time-saving tool.
However, regardless of how technologically advanced the Total Station is built, it takes a competent surveyor to ensure an accurate and precise instrument station set-up. Let’s take a look at the steps involved to properly set up and level the Trimble Total Station.
The survey equipment in this instructable can be purchased through the manufacturer at www.trimble.com. No endorsement intended.
Disclaimer – “This instructable has been created as a simple guide to set-up and level the Total Station surveying instrument. It should be noted that I am not a professionally licensed surveyor and this guide does not demonstrate specific surveying operations. Safety considerations include caution when moving the sharp legs of the tripod and appropriate clothing as the local climate dictates”.
Teachers! Did you use this instructable in your classroom?
Add a Teacher Note to share how you incorporated it into your lesson.
Step 1: Gather the Survey Equipment, Stake, and Tools.
It is very important for every surveyor to have all of their equipment ready for the job ahead. One missing item could cost your organization time and money.
Prepare the following items:
c. Total Station
d. Survey Controller
e. Power Cable
f. Communication Cable
g. Battery Pack
h. Survey Controller
i. Stake w/nail and survey marker
Step 2: Establish a New Temporary Benchmark.
Within our projects we typically need to establish a new point of reference with a temporary benchmark.
Using a small nail tapped into the center of a wooden stake, pound the stake into, and as flush as possible, with the ground.
Step 3: Set-up the Tripod.
While holding the tripod, loosen the tripod leg clamps and extend the tripod up to a height near your neck and chin. Tighten the leg clamps.
Spread out the tripod legs evenly, about two to three feet for each leg, and center it over your benchmark stake. The top of your tri-pod should be mostly level and parallel with a horizontal plane.
Step 4: Attach the Tribrach and Coarse Level the Tripod Over the Benchmark.
A level instrument is a key to successful surveys. The tribrach features a level bubble and leveling screws to adjust the tripod’s level as necessary. An optical plummet on the tribrach allows the surveyor to view the tribrach’s center and place its cross-hairs precisely over the survey marker nail on the stake.
Center the tribrach on the tripod and attach it via the tripod connecting screw.
While holding two tripod legs, keeping the tripod as level as possible, place the optical plummet cross-hairs over the center of the nail.
With your foot, apply equal pressure along each tripod leg to ensure a stable connection with the ground.
Step 5: Continue to Level and Adjust the Tribrach As Necessary
Viewing the level bubble, adjust the height of two tripod legs so that the bubble is close to center. Then use the tribrach leveling screws for a more precise bubble level.
Alternate back-and-forth between the optical plummet and level bubble to gain the most center and level position. You may need to loosen the center tripod screw to shift the tribrach for a more center cross-hair position.
Step 6: Set the Instrument on the Tripod.
Remove the Total Station from the carrying case and secure it to the top of the tribrach with the tribrach/instrument lock.
Be careful with this step as the instrument can cost up to $25,000! I know my bank account couldn’t support that bill.
Step 7: Connect Power Supply and Communication Cables.
Attach the battery pack to the tripod.
Connect one cable from the battery pack to the instrument connection port.
Connect the other cable from the battery to the survey controller.
Step 8: Power-on the Instrument and Controller to Access the Fine Level.
Press the black button on the instrument to power it on.
Press the green button on the survey controller to power it on.
On the survey controller, use the stylus to open the survey controller program. The software will open fine-leveling automatically.
Step 9: Fine Level the Instrument and Compensate.
To fine level the instrument, turn the total station so that its face-plate is parallel with two tribrach leveling screws. These two screws will be used to adjust the trunnion (left and right) and the sighting (forward and back).
Now that the instrument has been successfully leveled, press the accept button so that the instrument can perform its compensation action.
Step 10: You're Ready to Survey!
The total station is now ready to begin surveying operations. Have a blast!