Introduction: Arduino Ohm Meter
The simple principle behind this is the voltage divider network. The value of the unknown resistance is displayed on the 16*2 LCD display.
16 * 2 LCD display
Teachers! Did you use this instructable in your classroom?
Add a Teacher Note to share how you incorporated it into your lesson.
Step 1: About Project
The working of this Resistance Meter is very simple and can be described using a voltage divider network shown below.
From the voltage divider network of resistors R1 and R2, Vout = Vin * R2 / ( R1+R2 ) From the above equation, we can deduce the value of R2 as R2 = Vout * R1 / (Vin - Vout) Where R1 = known resistance R2 = Unknown resistance Vout = voltage at R2 with respect to ground Vin = voltage presented at the 5V pin of Arduino So if we see the value of voltage across unknown resistance (Vout), we can clearly determine the unknown resistance R2.
Here we have read the voltage value Vout using the analog pin A0 and converted those digital values (0 -1023) into voltage as explained in the Code. If the value of the known Resistance is higher or smaller than the unknown resistance the error will be more. So it is necessary to keep the known resistance value next to the unknown resistance.
Step 2: Run a Program
LiquidCrystal lcd(2,3,4,5,6,7); //rs,e,d4,d5,d6,d7
int Vin=5; //voltage at 5V pin of arduino
float Vout=0;//voltage at A0 pin of arduino
float R1=3300; //value of known resistance
float R2=0; //value of unknown resistance
lcd.print("R (ohm) = ");